We have seen issues after updating the firmware causes your Kestrel to disconnect frequently. The fix for this seems to be just restarting your mobile device. After restarting the mobile device, try connecting again to the new Kestrel LINK app... Permalink
Wind Speed – NK sells factory calibrated replacement impellers that can easily be installed by the customer. NK recommends replacing the impeller every 2-3 years depending on usage.
Humidity – The Kestrel 5 Series and DROP data loggers employ a highly accurate, calibrated, digital humidity sensor. Based on the sensor manufacturer's specifications and NK testing, drift on this sensor over time is minimal and field recalibration is likely to degrade accuracy rather than improve it. Factory calibration to validate and certify accuracy at a minimum of two RH points is available from NK. Kestrel 5000 series units can also be recalibrated in the field using our kit RH Recalibration Kit . During the five-year warranty period, NK will repair, recalibrate or replace any Kestrel 5 Series or DROP which drifts more than +/-1% RH/year since the date of manufacture.
Temperature – The thermistor in the Kestrel 5 series should not need to be recalibrated. If you think your sensor is incorrect, please contact [email protected]
Pressure – The pressure in the Kestrel 5 series should not need to be recalibrated. If you think your sensor is incorrect, please contact [email protected]... Permalink
Kestrel runs free Zoom training classes throughout the year on different topics. Check out the latest classes HERE
For live training at a shooting site, you can check out some of these optionsPermalink
Please try refreshing the cache on the webpage. To do this, press Ctrl-F5. If that does not work, then please try opening the order in an incognito window (Shift-Ctrl-n) . For MAC users on Safari, you can use [CMD] + [ALT] + [E]... Permalink
It will depend on if you want to integrate just the weather data or both the weather and ballistics data. If you need the ballistics (and/or weather) data, please send an email with a high level description of the project to Tech Support Email.
If you just need the weather data protocol, you can fill out our Technology Agreement found here: Kestrel Technology Agreement
Our article Understanding Pressure, Altitude and Density Altitude talks about Altitude and Barometric Pressure in great detail. The article also describes how to utilize your Kestrel to take these measurements.... Permalink
Every single Kestrel Meter and Kestrel DROP logger is factory tested to validate performance within specifications before it leaves our factory. We don’t rely on spot checks because we know you rely on your Kestrel to give you the accurate readings you need to stay safe, make decisions, and enjoy your activities. Each of our calibration standards is traceable to the National Institute of Standards and Technology (NIST), ensuring the highest level of reliability. Our wind/airflow standard is actually calibrated in the NIST laser doppler wind tunnel as this is the only lab in the country that can provide the calibration accuracy we require.
NK offers three types of calibration services to meet your accuracy validation needs. Each of these calibration services can be conveniently ordered through our website, allowing you to cross “Get my Kestrel calibrated” off your to-do list whenever convenient. And, ordering your calibration online creates an order history you can refer to in future. But, if you prefer, you can always call or chat with us during business hours to order your calibration. For more in-depth information about each of the services described below, simply click through to the relevant ordering page.
NIST Calibration is our most in-depth calibration service. The Kestrel’s measurements are compared to our NIST-traceable standard at two points after an appropriate equilibration period. The NIST Calibration Report and Certificate will show as found and as left values, standard values, acceptable range, and deviation. It will also list the date, serial number and model of the unit, desired recalibration interval, and test technician’s name and signature. NIST Calibration services can be added on to any new Kestrel purchase for customers who require it, or you may return your Kestrel to us at any time after purchase to validate its performance in detail. Please visit our Kestrel Calibration product for more information.
For Factory Calibration we repeat the tests performed on your unit when it was manufactured and provide a renewed Sensor Calibration Certificate confirming performance within specification, the serial number of the unit, and the signature of the testing agent. Please note that if you need more than one Factory Calibration, Tune-Ups offer a better value! Please visit our Kestrel Calibration product for more information.
Tune-Ups include Factory Calibrations of all sensors on your unit as well as a new impeller, new battery, functional checks, cleaning and leak testing. Tune-Ups are priced at a flat $75 per unit so offer a great value if you need more than one Factory Calibration or just want to ensure your Kestrel is fully tested.
Note on Sensor Correction: If needed and available for the sensor in question, we will also restore your unit’s performance to specification by either replacing the impeller or correcting the sensor. Although not correctable, in our experience, both the temperature sensor (ambient and globe) and newer humidity sensor in the DROP and Kestrel 5 Series have proven extremely stable in performance. Every Kestrel is guaranteed to perform within specification, including the specified drift of the sensor in question, for Five Years from the date of manufacture or documented date of first customer purchase. If your Kestrel is under warranty and does not meet this standard under test, it will be repaired or replaced as covered under the terms of our Limited Warranty. If your Kestrel has an unrepairable problem and is no longer under warranty, you have the option of trading it in for a new or upgraded model under our Kestrel Trade-in, Trade-up Program. Please visit our Kestrel Tune-up product for more information.
Shipping Your Unit: After ordering a calibration or tune-up service online or via phone, please note your order number so you can add it to the outside of the box. Please pack your Kestrel securely and ship to:
21 Creek Circle
Boothwyn, PA 19061
You will also receive an email confirming these instructions. It is most important that we have your Order Number so we can promptly match your unit with your service request. You can also print your order confirmation and add it to the box.
Questions or clarifications about our calibration services? Email
"All measurements not available in all units"
For best accuracy when taking temperature measurements you must keep the air moving around the temperature sensor and keep the Kestrel out of prolonged, direct sunlight. If there is a strong breeze, just point the Kestrel into the wind. If there is no wind, swing the Kestrel around in a circle on its lanyard or rapidly wave it back and forth in hand. This ensures that the sensor measures the temperature of the air rather than the temperature of the case. Alternatively, if you have time, allow the Kestrel to rest where you want to measure the air temperature (NOT in your hand or in direct sunlight if there is no wind) until the case temperature equalizes with the air temperature. When equalized, the display will read a constant value. Be aware that this could take a few minutes, particularly if there is no wind.
One reason your Kestrel may not be reading accurate ambient humidity is because of proximity to humidity sources such as your hand, your body or the ground. Avoid covering the large humidity chamber opening on the case rear with your fingers. Keep all fingers below the finger groove and away from the chamber opening. Do not take reading that rely upon humidity with the Kestrel meter lying flat – raise it into the air vertically or stand it on its base and orient it into the wind or air flow. If a high level of accuracy is required for measurements involving humidity, do not hold the unit while taking measurements. Taking readings with the Kestrel in your hand or over your body can raise relative humidity by 5% or more. A Kestrel vane mount will isolate the Kestrel Meter from erroneous humidity influences and keep it oriented into the wind or air flow while taking measurements. Whenever you move your Kestrel Meter into a new environment wait until the primary sensor values have stabilized (Temperature, Humidity, etc.) before taking measurements that rely upon these values. After a large change in environmental condition, this may take 15 minutes or more, with humidity potentially taking the longest. If the humidity sensor housing (located above the rear label) is wet, the Kestrel cannot take an accurate humidity reading. If the sensor does become wet, shake the unit vigorously to remove the water, and then let it dry completely before use. If the sensor comes in contact with salt water, rinse it thoroughly with clean water, then shake it and allow it to dry. As is typical with humidity sensors, it is possible for the humidity sensor in the Kestrel to drift over time as described in the spec sheet. If your unit has a Humidity recalibration function, it can be returned to the factory or used with the Kestrel RH Calibration Kit to recalibrate its humidity readings.
Wind speed is highly dependent on surrounding obstructions such as buildings, yourself and the ground, with wind speeding up as it flows around and over obstacles and slowing down as it passes behind them. Even in an open area, Wind speed will be slower near the ground so ensure you are taking measurements at an appropriate height for your application. Hold the back of the unit directly into the wind. Holding the unit off-axis by 5° will reduce accuracy by 1%, 10° will reduce accuracy by 2% and 15° will reduce accuracy by 3%. Using a Kestrel Vane Mount will ensure the unit is correctly oriented into the wind and measuring accurate wind speeds. The Kestrel impeller housing can rotate in the unit. For best accuracy, ensure one of the three “arms” of the impeller housing is pointing straight up. The wind speed displayed by the Kestrel Meter is a 3 second rolling average. This allows the unit to provide a value that is more representative of the typical, ambient conditions but not capture peak speeds for instantaneous blasts of air.
The compass in the Kestrel Meter is used for direction, crosswind, headwind/tailwind as well as for capturing various user inputs based on direction. When calibrating and when using the compass, it is important to hold the unit as vertically as possible. Any tilt in angle present when the unit is calibrated or when measurements are taken will negatively impact the accuracy of the compass reading. Many users will inadvertently tilt the Kestrel Meter away from themselves slightly when holding the unit. If a high level of accuracy is required, using a Kestrel vane mount on a leveled tripod for calibration and measurement can eliminate error.
All pressure and altitude measurements are made using the pressure sensor. Some units have a dedicated “Pressure” screen which shows Station Pressure, the raw pressure reading straight from the sensor. The Barometric Pressure measurement (Baro) displays the local Barometric Pressure using the Station Pressure measurement adjusted to the local altitude using the Reference Altitude value input by the user. The Altitude measurement displays the local altitude using the Station Pressure measurement combined with the Reference Baro value input by the user. To obtain an accurate barometric pressure or altitude readings, you must first know EITHER your location’s current barometric pressure OR your current altitude. If the Reference Altitude value is incorrect, the Barometric Pressure reading will also be incorrect. If the Reference Barometric pressure value is incorrect, the Altitude reading will also be incorrect. Because Barometric Pressure and Altitude are each dependent on the alternate reference value remaining constant, Barometric Pressure and Altitude cannot be accurately measured simultaneously. If your unit has a pressure recalibration function, a new Barometric Pressure calibration value may be input. If this value was not taken from a verified source (a local airport or weather station) with the Kestrel being calibrated at the same location as the reference device, pressure related measurements may be incorrect.
When changing environments (such as moving form an air conditioned room to outdoors or removing the unit from your pocket) the unit requires between 8-15 minutes to equilibrate to its surroundings before taking readings. Take measurements at least 3 feet off the ground and in the same wind or air flow conditions as the people you are monitoring. Ensure the Kestrel is oriented into the wind and able to measure the full wind value. A tripod or pole mount and the Kestrel Rotating Vane Mount are ideal for ensuring accurate measurements. Differences in the reflectivity of ground surfaces such as grass or asphalt will impact measurements. Be sure to take measurements in the same solar/radiant heat environment as the people you are monitoring. Unlike best practices for other measurements, WBGT is intended to be taken in direct sunlight. As long as there is occasional wind the Kestrel has software that allows it to maintain an accurate WBGT reading despite being placed in direct sunlight.
Kestrel Professional Meters can calculate air flow through a duct by combining user input information about the size and shape of the duct with the measured air velocity. In addition to reading the instantaneous Air Flow off the main measurement screen, a more accurate result may be obtained by capturing an average air flow on the Min/Ave/Max screen while traversing the duct. Traversing consists of dividing the area of the duct into evenly sized sections and averaging the air flows at the center of all sections. By spending an equal amount of time at each measurement location and quickly moving between measurement locations, the average air flow measurement can provide a more accurate measure of the air flow in the duct. Capturing an average can be particularly helpful when measuring ducts with registers or dimensionally uneven air flow.
(Kestrel 5200 Professional Environmental Meter) Kestrel Professional Meters can calculate a concrete evaporation rate (ACI 308) by combining a user input concrete temperature with measured air temperature, wind speed and relative humidity. Concrete mix temperature is generally measured with a probe or infrared thermometer and should be read at the time of making the evaporation rate measurement After inputting the mix temperature, hold the unit vertically, approximately 20 inches above the surface of the concrete while facing the rear of the unit directly into the wind. To avoid inaccurate Evaporation Rate readings due to thermal loading it is best to shade the Kestrel. (Be sure the source of shade is not obstructing the wind speed measurement.) The ACI recommends taking a 6-10 second average of the evaporation rate to account for fluctuating wind speeds. To measure an average Evaporation Rate, scroll right from the Evaporation Current Measurement Screen to the Min/Ave/Max Screen and press select to manually start and stop a capture.... Permalink
You will first need to create an account if you do not already have one. After you create an account, sign in. Click on your account name at the top of the page and select My Account. On the left of the screen, you will see Product Registration, select this option. Enter your purchase date, serial number and product name into the fields and then select Save.... Permalink